Descripción: manual electrocardiógrafo Cardiofax DECRETO LEGISLATIVO N° DEL REGIMEN DICIPLINARIO DE LA PNP (RESUMIDO). docx. D.L. DECRETO LEGISLATIVO QUE REGULA EL REGÍMEN DISCIPLINARIO DE LA PNP TITULO PRELIMINAR PRINCIPIOS. Grupo Ascenso PNP. LEY DEL RÉGIMEN DISCIPLINARIO DE LA PNP – DECRETO LEGISLATIVO Nº DEL 11 DE DICIEMBRE DE
|Published (Last):||17 August 2005|
|PDF File Size:||1.74 Mb|
|ePub File Size:||13.45 Mb|
|Price:||Free* [*Free Regsitration Required]|
Select the desired item with the or function key.
EQWEQ by Siete Tacsi Avila on Prezi
The lifetime of the battery is about 6 years. When a narrower recording paper is used and the paper detection circuit cannot detect it, the thermal head may be activated.
If necessary, provide additional protective earthing, which is permanently connected see also Push the paper magazine release button and open the paper magazine. Mount the electrocardiograph by aligning the holes on the rear panel over the set pins.
Insert the SD memory card into the SD card slot on the right side panel of the electrocardiograph before saving the system settings. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people. CAL mark The height corresponds to 1 mV waveform amplitude.
Decreti the Surface of the Electrocardiograph Clean the outside surface of the electrocardiograph with a soft cloth moistened with water, neutral soap or alcohol Cleaning the Sensor The paper empty sensor is located as shown left. Retry saving the ECG data files or check the media. Check that the recording paper is set correctly.
Do not charge a deteriorated battery pack. To remove the SD memory card: Recording result is partially missing or When the thermal head is touched with a dirty hand, or comes in faint. The impedance of the protective earth increases and the patient and operator may receive electrical shock or injury. Avoid vibration and mechanical shock, even during transport. Lit in orange Battery charging lamp: CAUTION The electrocardiograph should not share the power line with any instrument which consumes a lot of power, such as an X-ray instrument, because it may cause artifact.
Select the file and press the Delete function key. Select whether to display the age on the Patient Information screen.
Manual Cardiofax – PDF Free Download
When the instrument is not functioning properly, it should be clearly marked to avoid operation while it is out of order. The settings are not changed and previous screen appears.
Lit Battery power lamp: Changes the character type. This is used to check the memory by comparing the data of the test patterns written to and read from each memory area.
decreto legislativo 1150 pdf
These cause malfunction and shorten the life of the electrocardiograph. If on either AC or battery operation. Additional protective earth or separating transformer For B: In addition, service, technical modification or any other product change performed dcereto someone other than NKC or its authorized agents without prior consent of NKC may be cause for voiding this warranty. In this case, the electrocardiograph stops printing to protect the thermal head. Patient names on the screen shots and recording examples in this manual are fictional and any resemblance to any person living or dead is purely coincidental.
Failure to follow this instruction may cause the electrocardiograph to fall over or cause battery liquid leakage. If a new and old electrode, or a reusable and disposable electrode are used together, the ECG waveform cannot be recorded due to high polarization voltage. The characters are toggled. Examples of color-fading materials made of soft vinyl chloride: The cursor moves to the gender column.
For details, refer to the Service Manual. In manual recording mode, the baseline is automatically adjusted every time recording is started and the lead group or sensitivity can be changed. If the liquid inside the battery contacts the eyes, mouth or skin, do drcreto rub the eyes, mouth or skin.
When operating the electrocardiograph with the battery pack, check the date when the battery pack was last replaced on the start date label attached to the electrocardiograph.
If the memory is not faulty, power loss might be caused by static electricity. Clean the display using a dry soft cloth or derceto cloth which is moistened with neutral soap and wrung out. Avoid direct sprinkling, spray or moist air from a nebulizer or humidifier.
To return to the Data Management screen, press the Back function key. There is a time delay between the input signal and output signal. Effect of direct or indirect electrostatic discharge: To turn on the backlight, press any key.
Repeat steps 8 and 9 to change other settings. Failure to follow this warning may cause electrical shock or injury. Search for the character in the list on the next two pages. Defective products or parts must be returned to NKC or its authorized agents, along with an explanation of the failure.
The Data Management – Format legislatvio appears. Recorder cover release button 2. Off Battery operation lamp: